Influence of double rods and interbody cages on quasistatic range of motion of the spine after lumbopelvic instrumentation
TypeArticles dans des revues avec comité de lecture
Purpose This in vitro biomechanical study compares residual lumbar range of motion (ROM) and rod strain after lumbopelvic instrumentation using 2 rods, 4 rods and interbody cages. Methods Seven human cadaveric specimens were instrumented from L1 to sacrum, and pelvic screws were implanted. The pelvis was constrained and moments up to 7.5 Nm were applied to T12. Segmental L1–S1 ROM was analyzed by tracking radiopaque balls implanted in each vertebra using biplanar radiographs. Deformation within principal rods was measured by strain gauges. Four configurations were compared: 2 rods (2R), 4 rods (4R), 4 rods + ALIF at L4–L5 and L5–S1 (4R + ALIF), 2 rods + ALIF (2R + ALIF). Results Intact average global L1–S1 ROM was 42.9° (27.9°–66.0°) in flexion–extension (FE), 35.2° (26.8°–51.8°) in lateral bending (LB), 18.6° (6.7°–47.8°) in axial rotation (AR). In FE, average ROM was 1.9° with both 4-rod configurations versus 2.5° with 2R and 2.8° with 2R + ALIF (p < 0.05). In LB, ROM ranged between 1.2° and 1.5° without significant differences. In AR, ROM was 2.5° with both 4-rod configurations versus 2.9° with 2R (p = 0.07) and 3.1° with 2R ALIF (p = 0.01). In FE, strain decreased by 64% and 65% in principal rods at L3–L4 with 4-rod. When comparing 2-rod configurations, strain decreased by 1% in flexion and increased by 22% in extension at L3–L4 when adding an ALIF at L4–L5 and L5–S1. Conclusions Double rods and interbody cages decrease residual ROM in FE and AT. Double rods seem efficient in limiting strain in principal rods. The use of single rods with cages at the lumbosacral junction increases strain at the first adjacent level without cage.
Files in this item
Showing items related by title, author, creator and subject.
CHARLES, Yann-Philippe; PERSOHN, Sylvain; ROUCH, Philippe; STEIB, Jean-Paul; SAULEAU, E.A.; SKALLI, Wafa (Elsevier BV, 2014)Background : Lumbar stenosis and facet osteoarthritis represent indications for decompression and instrumentation. It is unclear if degenerative spondylolisthesis grade I with a remaining disc height could be an indication ...
Clinical Outcomes and Complications After Pedicle-anchored Dynamic or Hybrid Lumbar Spine Stabilization : A Systematic Literature Review PRUD'HOMME, Marion; BARRIOS, Carlos; ROUCH, Philippe; CHARLES, Yann-Philippe; STEIB, Jean-Paul; SKALLI, Wafa (Wolters Kluwer, 2015)Fusion is the standard in-strumentation for many pathologies of the lumbar spine. Wor-rying rates of failure, including adjacent segment degeneration (ASD), have consistently been reported. The interest for dy-namic ...
Limiting interpedicular screw displacement increases shear forces in screws: A finite element study LIMA, Lucas Venancio; CHARLES, Yann-Philippe; ROUCH, Philippe; SKALLI, Wafa (Elsevier, 2017)Background context: Screw loosening has been reported for non-fusion devices. Forces on pedicle screwscould be related to kinematic parameters as the interpedicular displacement (ID), which consists of thedisplacement ...
LOISEL, François; DURAND, Stan; PERSOHN, Sylvain; AUBRY, Sébastien; LEPAGE, Daniel; BONNET, Xavier; SKALLI, Wafa (ELSEVIER, 2020)The scapholunate joint is one of the keystones of the wrist kinematics, and its study is difficult due to the carpal bones size and the richness of surrounding ligaments. We propose a new method of quantitative assessment ...
Relevance of using a compressive preload in the cervical spine : an experimental and numerical simulating investigation BARREY, Cédric; ROUSSEAU, Marc-Antoine; PERSOHN, Sylvain; CAMPANA, Sophie; PERRIN, Gilles; SKALLI, Wafa (Springer, 2015)Simulating compressive action of muscles, a follower load attends to reproduce a more physiological biomechanical behaviour of the cervical spine. Only few experimental studies reported its influence on kinematics and ...